MegaPhase GrooveTube® cables have successfully contributed to the first and only asteroid dust capture from space in the Hayabusa, formerly known as MUSES-C Space Program. The Hayabusa seven-year long space program was equipped with MegaPhase’s GrooveTube® cable assemblies. (more…)
Archive for the ‘MegaPhase Insight’ Category
Our customers often see an increased involvement of top management because of the responsibilities called out in our Quality Management System (QMS), ISO 9001:2008. This starts with our Quality Policy, Goals and Objectives. It continues with a Management Review of data generated by our QMS. Taking actions to make sure that goals are met, new goals are set, and continual improvement is achieved. (more…)
GrooveTube® is another name for the MegaPhase super-flexible outer conductor used in many of our cable assemblies including: Warrior, GrooveTube, RF Orange, RF Green, VNA, SuperFlex, SiteLine, T-Vac Series, and BT Series. (more…)
Contact Pins and Dielectrics can be damaged if misaligned connectors are mated. Make sure that mating interfaces are parallel and on center during Mate/De-Mate cycle
Plating and surface finish of outer and inner contacts can be damaged if connector bodies are allowed to rotate during Mate/De-Mate cycle. Use wrench flats on connector bodies to keep them stationary while rotating coupling nut during mate/de-mate.
Minimized Load Configuration
Allow test cable to assume natural bend configuration route between test port and device port. Loosen test port connection when test configuration is largely different than test configuration. Route the cable, connect device port loosely then tighten both port connections to recommended coupling torque. This method will minimize most bend and twist loading issues.
We have essentially learned thus far that we need to properly care and handle cable interface interconnects, interface cleanliness, interface gauging, and recommended coupling torque. Another important aspect to cable care handling is cable routing:preventing over-bending!